HARON, S., NABOK, A. V. and RAY, A. K. (2003). Optical biosensing transducer based on silicon waveguide structure coated with polyelectrolyte nano layers. In: Conference on Bioengineered and Bioinspired Systems, Maspalomas, Spain, May 19-21, 2003. 100-108.Full text not available from this repository.
An optical biosensor based on attenuation of the light intensity during multiple reflections. in a planar waveguide has been developed for water pollution monitoring. The planar waveguide consists of a 190 nm thick silicon nitride (Si3N4) core layer sandwiched between 1.5 mum thick silicon dioxide (SiO2) cladding layers. Composite polyelectrolyte self-assembled membranes containing Cyclotetrachromotropylene (CTCT) as an organic indicator and enzymes, such as Urease or Acetylcholine Esterase (AChE) were deposited on top of silicon nitride core layer within a 4 x 6 mm sensing window. Experimental studies on the light propagation through the planar waveguide show the advantages of this method over conventional UV-visible absorption spectroscopy, It was found that the planar waveguide sensitivity is higher by. several orders of magnitude than that for the UV-visible absorption spectroscopy. The respective enzyme reactions as well as their inhibition by heavy metal ions were studied by monitoring the light intensity in the planar waveguide. Cadmium (Cd2+) and lead (Ph2+,) ions were registered in very low concentrations down to I ppb with the planar waveguide transducer. The enzymes used were inhibited differently by the above pollutants, which is promising for the development of enzyme sensor arrays.
|Item Type:||Conference or Workshop Item (Paper)|
|Research Institute, Centre or Group:||Materials and Engineering Research Institute > Thin Films Research Centre > Electronic Materials and Sensors Research Group|
|Depositing User:||Ann Betterton|
|Date Deposited:||20 Apr 2010 17:12|
|Last Modified:||20 Apr 2010 17:12|
Actions (login required)
Downloads per month over past year